Click Image for Gallery
Thank you for choosing our brand-new Elitech refrigerant scale. To use the scale correctly, please read over the manual first, especially the "safety notice". It is suggested to keep the manual properly with the scale or in a place where you can refer to it conveniently.
Max weighing: 110kg(220Lb)
Temperature fluctuation range:5℃/h
Power supply: 5AAA batteries
The maximum weight of the scale is 100kg.
1.Take out the scale from the package and place it on firm and level surface
2.Draw out the spiral line of the hand-held device from the side of the scale
3.Press the "power"button on the hand-held device,the buzzer will beep once indicating self-test is finished.
4.Press"unit"to switch between "Kg","Oz"and"Lb".
5.Press"zero"to zero(tare)the scale
6.Place the refrigerant tank in the center of the scale platform ;zero (tare) it again to display the weight of refrigerant.
7.Observe the change of refrigerant weight real time during charging and recovery.
over range indication
1.Power-on self-test:it lasts less than 2s.
2.Tare deduction in full range(100kg)
3.The LCD displays numbers from high digit to low digit.The digit that exceeds the digit range will not be displayed
4.Auto off:If the weight on the platform does not change for 10 minutes,the scale turns off automatically.
1.Please use the scale in specified environment to avoid error or damage of the sensor.
2.Please do not use chemical solvent or corrosive liquid to clean the scale.
3.Please do not disassemble the scale body and inner device;otherwise,it may cause great weighing error.
4.Please do not pull the spiral line hard in case measuring error is enlarged or the sensor damaged
5.Remove the batteries if the scale is not to be used foe a long time in case the leakage corrodes the scale.The batteries included are not rechargeable.Please do not charge them.
LMC-300 Electronic Refrigerant Scale(new product)